We have multiple DT80M datalogger over the country and for some reason, they many of them have started to lost the SIM pin code. This affects only for this settings, all others ones are fine. It happens randomly, between month or like once in a year. After entering the SIM Pin again, everythings works like excepted. This phenomen was not going on a couple of years ago with these logger
Hello, Following our system OFF/ON, the integrated DT80 has lost its configuration (no current job). This problem has already been seen on other DT80 that we use. The DT80 had already "survived" previous ON/OFF, the data loss appears random.
Moreover, after the configuration loss it has been then impossible to reload the configuration of the DT80, even after freeing disk space.
I'm using series3 DT80. After firmware upgrade to last version v9.22, logger was working fine for a week, untill I wanted to clear logger internal memory using command DELALLJOBS. Result of "delalljobs" command was that current job was terminated but message SW exception appeared. Moreover, internal disk was not cleared.
Now, logger display show message "No current job", but I
To collect data from a serial device we use manual channels. There are some problems to realize it with dEX-2:
The available space to enter commands is limited. (We could solve it by splitting it to several channels)
After saving the configuration to a file and reloading it, there are suddenly many blank lines inserted that we did not originally enter. After each save, this seems to double
Hi Facing an Issue with DT82i connectivity. I was able to connect device and create channels and even sent data using Modbus TCP, but all of sudden the status it is showing is "device is currently offline" in dEX 2.1.18 application and even web interface is not working its just loading. As in Device LCD screen it was showing message "SW Exception" but after deleting all jobs from the device
When using dEX-2 to apply a schedule to our dataloggers, the dataloggers now give the error "SW Exception".
Some times the schedule does no load and there is "no config", sometimes it does load. But we get the error code every time we try to apply the schedule. The dataloggers and dEX 2 were working fine for two weeks but then this issue started. At first it was every second time but
Good Day Everyone, I am facing problem with applying program to logger since the power always changes from setted 'always' to 'only when taking reading' which causes our equipment to not function accordingly.
While troubleshooting a strange voltage reading on a DT85 S3, I narrowed it down to this. When I measured the reading outside of a schedule, it measured the voltage correctly. When the same line of code was used in a schedule, it measured the incorrect value.
Actual voltage measured with a Fluke multimeter at the DT85
terminals: 44.7 mV
Voltage manually measured by DT85 in DeTransfer without
I have a series 3 DT85 v9.14. It as not been used much over the last year, and now just when I need it it seems to be giving me an error. As it collects data it will, all of a sudden, stop, lights flash, and the message SW exception appears.
I was having a problem with the unit running on battery, in that it was losing power then turning back on then losing power. I traced that t
I need to measure an analog voltage of +/- 5 Vdc. I know I need to set the channel 10:1 attenuator in order to do this. When I configure the input range to '30V Input' on my channel in dEX and save to the logger, I get the error message as seen in the attachment.
I've tried doing the command through the command window in dEX also [9HV(GL30V)]. Then it seems to be accepted, but when I apply a
I have been experimenting using a DT85 Series 2 to measure relative humidity using the Wet/Dry bulb method. I have been taking the measurements using a pair of K type thermocouples - one of which is 'wet.'
I have tried using the routine outline in the FAQ through DeTransfer. The problem is that the calculation in the formula is quite different from that using a psychrometric cha
I want to run the DT85 via USB, so therefore I need a USB-Driver. Unfortunately, the Driver on the page seems to be missing. Also, the dEX software platform, which I thought I could find is leading me to an "not found" page. Could you help me with that problems?
I noticed in the Release Notes for a recent logger purchased that has the 9.10 firmware installed that a new running average channel option has been implemented. I thought this would be useful, but I can't seem to get it to work.
I created a manual channel in the configuration builder "Configure the logger", and even when I copy verbatim the channel example in the release notes, when
I have got a problem regarding the modem power config. I have fully configured a DT80M (system name, time and so on) including the modem power config, which is set to "ALWAYS ON" due the need of the DT to act as a http server. If I make a HARDRESET or a power loss occurs, the system reloads the config correctly, all parameters are set as its config value except the modem power config,