posted Jan 31 at 9:00 am
We have a DT800 since many years and we use it a few times for temperature measurings
Now, i have a new chalenge to build a weather station with some sensores the we have in the lab
The sensors are:
SP Lite2 Pyranometer; sensitivity 68 uV/W.m-2
Thies anemometer 0.5-40m/s, a T351-TR Thermistor, a very old pluviometer that I will calibrate it.
And some k thermocouples
Hi, I have a couple of DataTaker DT80's (one series 2, one series 4).
I plugged the S4 into a Windows 10 PC and Windows said it was installing a driver. Would this be DTusb or do I need to physically download it (where from?) and install it first?
I see from other posts that DEX 2.0 avoids the need for Flash. Where is the download for DEX2.0? Do I need to upgrade the DT80 firmware firs
Hi I have a DataTaker 605 Series 3 and I do not have the software CD is there somewhere online this can be found? it says in the manual the software is provided free with each unit.
Any help would be wonderful.
I goofed today, and set the logging interval much shorter than usual (2 vs my usual 15 or 30 seconds), ran out of memory half way through my trial and overwrote the early data.
I have downloaded and dumped the DDB - it's still missing the first two hours or so, so I'm fairly sure they are gone.
I was watching the data in a DEX2 dashboard while it was being collected - long shot
My DT80 series 2 just stopped responding
The sample light, disk and power lights are on but no display
It was a brand new logger (2009?) . I had updated to latest firmware.
It was working fine for about 3 weeks
Anyone got any ideas ? A hard reset has not fixed it
posted Apr 14 '19 at 11:35 am
I'm currently evaluating a DT80M Datataker Logger for the use with our SDI12 Moisture sensors. Logging the data works fine so far but I hit a strange problem uploading the data to an FTP server. Every time the Logger tries to upload data, it fails with some strange 'FTP: server out of sync (-3)' error.
DT80> SESSION: Starting (manual)
SESSION: Ready (MEDIONmobi,3G-HSPA,-93dB
posted Jan 30 at 5:59 pm
Hi all, I last set up my DT85 on dex-2 about six months ago and have unfortunately forgotten how to do it. I am sure that the IP address and port 80 are fine but think there is something to do with the template that I am missing? Could any please provide me with the chronological steps I need to do as at the moment it keeps saying my DT85 is offline.
One of our DT85 stop working yesterday.
The display is working(has display when bootstrap) but nothing display.
Sample, disk and power LED is On.
Hard reboot is not working too.
posted Dec 5 '22 at 1:56 am
cách lập trình và so sánh giá trị trigger cho relay trên DT82E bằng phần mềm DEX-2
Ví dụ: Tôi muốn đo một bộ nhiệt độ từ 0 độ C đến 1000 độ C. Tôi muốn lấy bất kỳ giá trị nào n
posted Nov 14 '22 at 9:32 am
In order to drive cooling fans for my system, I need a dynamic voltage output (2 different values) depending on measured temperature.
I have a DT80 series 4 logging temperature values.
I was wondering if the DAC could achieves this function using alarms?
I was only able to set the dac output value with manual commands, and the doculmentation is not really exhaustive on how to driv
I'm sure this is a simple question, but I can't find the answer!
I want to establish a direct connection to the command interface over a local network using my software. The final aim is to have scripts running on a linux web server retreiving real time measurements from the DT80.
The DT80 is happily running on a local network, and there is no problem connecting with deTransfer and other ut
posted Oct 20 '22 at 1:25 pm
I just started using the DT80 series 4 and when I tried to connect the logger and it have been only showing "Connecting to logger.....please wait"
-I've rebooted the device,
-Connected it locally by the usb
-Configured a new IP address to connect it virtually
-I used firefox mozille to open the virtual page and it only shows "connecting to device.. please wait" the DT80 is
posted Oct 18 '22 at 5:09 am
We currently have DT80G devices in the field that upload to a FTP server. Is there a way to use a more secure upload channel?
I am still wrapping my head around this but there appears to be devices that can connect to MODBUS and could then handle the uploads. But I am not sure how all this stuff works and if I even technically understand how these things work
Dear DataTaker experts,
I am using DT85G Series 3, and before 2020, It was worked with dEX previous version without any issue. However, as it no longer work with flash player now I have to upgrade it to dEX 2.0 and firmware also upgraded to version 9.22
I have connected the logger using Ethernet cable and add the logger to dEX-2.
Now I cannot access the logger through the dEX-2 d
posted Sep 29 '22 at 7:25 am
I measure wrong voltages at any analog input.
The measured values are only 83% of the actual voltage applied.
Please find the result of the self test here:
Is there anything I can do?
How can I send the logger for repair?
posted Sep 2 '22 at 11:36 am
I have a requirement whereby i will have 2 DT85s in separate buildings both on different networks but pc has access to both networks, that i want to have 1 dt85 connected to the other or someway of having one download where all channels from both logger be on the one download.
I have data taker DT85W. I was heat up and now showing REFT error. How it can be rectified. Do I need to send it for repair or not
posted Aug 24 '22 at 6:23 pm
I have a DT85G-3 device connected by ethernet to a Windows Server 2016 PC. I am using dEX-2 to communicate with the device.
The device will periodically connect, show me the clock time and show me its data if I go to the Data Explorer. I can see that it is running Software Version 9.20.8942.
In particular I am trying to see the Schedules loaded on the device, which seem to be getting sent
I am logging 2 Pressure transducers (mV) and 2 digital counters on a DT80 series 2.
I have set the scan rate to 200mS. I have found that the DT
on about every 7th scan there is about 400-500mS interval
I know these are not high speed loggers but are there any settings I could use to speed up the readings. I only need to scan at the fast rate for about 20seconds during testing
Good day, can you assist please, i installed my DT80 in a truck to record data. Logger stopped recording, i set limit at 30Mb and programmed to stop recording when limit is reached. When i tried to download data through Dex 2, logger just hanged, did not download. I tried USB, download started but when i pushed X to light up the screen to check status, download stopped and setup cleared. I urgent