Hi, I'm pretty new to Dex2 and DT and I'm trying to connect a power meter using Modbus RTU (RS 485) to Serial Port in my DT85. I've tried to write my own script with help of many code examples found in internet which didn't work. I also tried to program DT with Dex2 using graphical features it provides.
Hi i am looking to save to a CV channel the time stamp of when temperature sensor connected to the DT85 reaches a certain temperature 75C. therefore the CV channel will merely have the time this occurred
I have somehow locked myself out of my DT80M logger. In the Dex 2 software I set user level security to on by mistake. Now when I try to connect to the logger from DEX I get asked for a user name and password. I have guessed the default as datataker and datataker as defaults without success. How do I reset this I cannot see an obvious way.
Hi I am using datataker Dt85. I am using web browser for configuring it Whenever I change the schedule and download it , the previous logged data is deleted Can anyone suggest a workaround. Should I use DEX instead of browser
Hello, I am a new user of DT85GM (and CEM20). I am going to use DT85GM for performing load test wherein we plan to use vibrating wire strain gauges and LVDTs (load cell is planned to be connected to another data logger). Since my background is in Civil Engineering so going through all the details of the user manual appears to be a time consuming task, something that our project timelines d
We are using the DT85 to track temperature data by the second with both k-type TC's and RTD's. We are converting the C degree to F with Polynomial 1. What we are finding when we export our data to CSV is that the temperatures have 6 decimal places. Is there a way to round these up/down before exporting?
We have just started using our DT85. It was working great in testing but now we have an issue. We are getting a DXC Compilation Failure when we connect with dEX-2. What could cause this and how could we remedy this? And now we can only connect 1 instance of dEX-2 at a time. Trying to connect on other devices at the same time causes the DT85 to shut down and reboot - then only one connects to it a
I have several DT80 running in radiometric stations. Now, I need to make modifications to the program of one the stations and seems to me I have to do it from scratch because I cannot modify the old program (which is very long...)
I want to program this part in the new "language": T(w,=1CV)
I have tried to do a manual to insert it. However, when I check the value of 1CV i
I have multiple DT85 devices (both 3 and 4). One of the series 3 units can be seen by DEX2 using either USB or ethernet connection but when I try and check connectivity it always says "Device Offline". It changes the light indicator on DEX2 from red to yellow.
I saw another post that said to check the console errors. Unfortunately, I do not have any console errors as DEX is seeing that there
Hello, I am having an issue where the DT85M occasionally stops logging and the commands stop working. When I try to send commands I get a "Permission Denied" response, even though I am logged in as an admin. I am getting a lot of E9, E10 and E12 errors. From a previous forum post it looks like it could be caused by commands leaked through the network. However, I have commands password prote
I use a DT82i with one device connected on rs232 host port. I retrieve the data frame from this device by decoding on a manual channel. Some values are sometimes negative due to the background measurement. I record the average of these values on CV every 10 min. I would like eject the neg values, how may I proceed ? Thank you
There is no session time window for ethernet connections. Is it possible to wake up the datalogger for certain time windows, in order to access it remotely over ethernet? We have a battery powered site with a radio link. The datalogger is connected to the radio modem over ethernet. In order to safe power, the datalogger stays in a low power mode and wakes up every day once to do measurements and
I have been trying to connect a DT85 series 3 logger to a windows 10 laptop for over a month, and am still having the same issues. Even with the most up-to-date deX2 and DT-USB software, my laptop is not recognising the logger when it is connected, and therefore, isn't opening the browser or letting me access any data.
It has been suggested that a firmware update (from 9.18
I've got a DT85 Series 2 with the latest 9.22 firmware and am configuring it with dEX-2 v2.1.25 but when I have it reading a couple of voltage channels I'm hearing internal relays clicking each time, is this normal? Although the expected values are correct, the noise is annoying. Am I missing a setting somewhere?
If its relevant, I'm making readings from pins 2 & 3 from the 4-pin connect
Hi, I had dex2 and delogger installed and running fine on a previous laptop with win10. I have just upgraded to a new surface pro, and I am getting an error for the usb driver . In device manager the following info is displayed for the DT8x datalogger
Windows cannot load the device driver for this hardware. The driver may be corrupted or missing. (Code 39)
Ozone Wallet: An Offline Wallet with Special Features Ozone is a scientific group including cryptocurrency experts who present an Offline Wallet in order to ensure traders and investors’ security. As is known, in a cold wallet, the user is not completely disconnected from the internet and in some cases, a World Wide Web connection is required for setting up. Hackers are aware of this mat