Recently i notice and read that to be able to see the configuration of your DT80 on the web is to migrate it in DEX2. Does it mean you will not be able to open the web portal since Adobe Flash is already EOL? Besides using DEX2 what other alternative can we use if we want to access the GUI on Web portal as we are accessing the dt80 via internet. Hope for your kind response. Thanks
i have problem when loading a program from a saved file.everytime i load the program,it will alter the calculation output(result) that will cause error when uploading the program to datataker.so everytime i want to resend the program to datataker,i need to re-do the calculation one by one. hope for a solution for this .thank you
We have to do auto shunt calibration process through program. For example, If we have Strain-1, then after offset removing process as we connect the 17134 Ohm or 174650 Ohm then strain value must come as 10000 or 1000 macrostrain, but unfortunately there again some offset for individual channel offset id differ from each other, So we want to remove this offset , Please suggest and suppor
I've been using a DT85 series 4 for logging several 4:20mA signals, and I'm configuring the DT-85 using DEX-2. the DT85 had displayed all the signal correctly on its display, however when I tried sending a CSV file using FTP, nothing was sent to the FTP server.
I am monitoring an air conditioning system for occasional trips of the compressor overload. While monitoring all the usual things like temperatures and pressures and compressor current at 30 second intervals, I also want to monitor the three compressor current transducers "as fast as possible" for 2 seconds following the contactor closure. I've looked at scanning fast in the manual (under "Contin
Hi, I use DT80W-S4 to measure analog (4-20mA) Water Level Sensor. When I calculate the mA that DataTaker measure is different from what the sensor output mA. How to make DT measure the same mA value as what sensor output? I did adjust the ohm of the 100ohm resistor but the resistor value only can be input round number.
Hello i habe trouble with my Programm on my DT85M Actually it is my first Project with a Datataker
I need to read 200 soil moisture sensors. Each sensor has 4 variables to measure 175 of them are with modbus where i have to recalculate 2 of the values 25 are with SDI12 The Sensors work and i can communicate with all of them. So i have 1150 values to measure with this sensor
While troubleshooting a strange voltage reading on a DT85 S3, I narrowed it down to this. When I measured the reading outside of a schedule, it measured the voltage correctly. When the same line of code was used in a schedule, it measured the incorrect value.
Actual voltage measured with a Fluke multimeter at the DT85
terminals: 44.7 mV
Voltage manually measured by DT85 in DeTransfer without
Adobe Flash Player's end of life date is set to 31 december 2020. In my University, we rely on it in order to use dEx. I would like to know what solutions datataker is providing to adress this issue ? I've lookep this forums's threads but couldn't find any about this.
Can the Ethernet Port on the DT85 be used to send a Modbus command to a TCP/IP Modbus Server Device (e.g. Sensor configured as TCP/IP Modbus Server) as well as receive a Modbus command from the TCP/IP Modbus Client Device (e.g. SCADA) simultaneously?
hi all, I am having difficulty being able to log onto the Data Taker I have, I have downloaded the dEX2 and the DtUsb software. The computer sees the Data Taker as it shows up in the devices tree, and when I connect with the USB I can see the IP. 127.1.1.1, but again I am unable to log into the Data Taker .
Hi! Noob here. I'm using my DT80 for simple temperature measurements, succesfully getting two simultaneous signals in ports 1 & 2. My issue is that I want to trigger logging schedules, i.e. by having a circuit in the upper left block (the 1D 2D...GND block) that will start the logging. How would i code this in the interface?
Thanks for any help, and sorry if this is unclear! Not fluent i
When i use the run wizard to set my mains frequency to 60HZ and temperature units to Fahrenheit , I continue to the end where i click on submit. When i collect my data it is in Celsius and when i check logger regional settings i see 50HZ and Celsius. I am using a DT85 with v9.22 firmware.